There are two common focusing methods:
Focusing step is placed before the deflection of the galvanometer, which can form a larger marking format
Dynamic post focus.The laser generates the laser and incident it to the vibrator deflector lens, and forms our preset marking pattern path with the control software. The vibrator lens refracting the laser to the focusing lens, and focusing energy to the focal length surface to complete the laser engraving.
To find the focusing method:
1.Continuous output test method
In marking paint a software about a centimeter square or round, conventional after filling, laser energy is transferred to set parameters column is bigger, use frequency, and then check the continuous marking, put a piece of metal in the product surface laser projection position card obstructions, such as continuous light marking, shake the Z axis movements, until the function to the strongest metal business card energy laser, the most clear voice, the brightest colors, is basically the focal length, lead the test a few times more, find the right focal length.This method is suitable for 1064 - nm wavelength of the optical fiber laser marking machine, semiconductor laser marking machine, end face pump laser marking machine, 355 nm laser marking machine, uv 532 nm green laser marking machine, etc., co2 laser marking machine can look for a piece of blank paper, and then draw a box in the marking software, continuous light shake the Z axis, hit the paper on the line for the focal plane when the finest.
2.Measuring focal length
One end of the steel ruler can be placed directly on the surface of the product.The scale value refers to a certain reference surface on the field mirror and can be shaken up and down to the scale data.This method is suitable for all laser marking machines, but has the disadvantage that it is not very practical if the product required for the radium carving is in a concave position or does not fit properly into a steel rule.
3.Double red light to point method
Add one or two oblique red light rays to the galvanometer or beside the galvanometer. Using some principles of right triangle, find another right Angle side on the plane of focal length by using a fixed right Angle side and two overlapping hypotenuse.You can quickly find the focal length by shaking it up and down until the two red indicators coincide.Although this method is convenient and fast, it requires the equipment manufacturer to cooperate with the secondary device. In addition, if the device is not properly adjusted or shifted in the middle, it is easy to cause misdirection and form the wrong focal length surface.